to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||21 June 2016|
|PDF File Size:||20.34 Mb|
|ePub File Size:||4.60 Mb|
|Price:||Free* [*Free Regsitration Required]|
Instrument settings The date and time as well as the set symbol accutrwnd appear on the display. Performing a measurement Blood application in the instrument: Performing a measurement The flashing drop symbol above the strip symbol now instructs you to apply the blood. Use only the following items for cleaning: The code strip is required at least when a new test strip container is opened and before the strips are used.
Performing a measurement Switching on the instrument Place the instrument on a level surface or hold it in your hand.
Memory Memory The Accutrend Plus instrument has two memory areas, each of which can be used to store up to measured values together with date, time and event flags. If the beeper is enabled, you will hear a beep in the following situations: Make sure the code strip belonging to these test strips is at hand unless the instrument has already been coded with this code strip.
Note that all handling of blood samples represents a risk of infection. Display Of Results Quality control recommendation Display of results When measurement is finished, the result is displayed. The code number last stored or used is displayed after powering on the instrument.
Cobas Accutrend Plus User Manual
Instrument settings Setting the date With the next three settings, you first enter the year, then the month and finally the day. When you press the button, the place of the measured value within the memory is displayed here: Instrument settings The default day flashes. Display Of Results Performing a measurement Display of results When measurement is finished, the result is displayed. Consult instructions for use. Page 81 Quality control recommendation What you need: The mem symbol indicates that the instrument is in memory mode.
Performing a measurement Check that the whole display is functioning correctly to prevent misinterpretations due to defective display elements.
Page 96 Memory Special cases: To run a test, an unused test strip is taken from the container and inserted into the instrument. Performing a measurement The three-digit code number which is also printed on the reverse side of the code strip and on the test strip container is shown in the display.
The display then automatically switches to the mode for setting the current day. From the barcode on the reverse side, the instrument detects which test parameter is to be measured and which code is needed for the test strip.
Performing A Measurement Performing a measurement Performing a measurement After it is switched on and coded, the instrument expects you to insert the test strip. Check that the whole display is functioning correctly to prevent misinterpretations due kanual defective display elements.
Performing a measurement Recommendations for the collection and measurement of capillary blood To obtain a suitable blood drop: Operating the instrument Replace and close the battery compartment lid. Quality control recommendation Preparing to perform a control test Get the test strip container for the required measurement e. Flap open displayed after instrument has been powered on The measurement chamber flap is not closed properly.
Contents Product specifications Operating conditions and technical data Performing a measurement Code strips The code strip provides the instrument with important information on the specific properties of the respective test strip. Press the Set button on the left side of the instrument to power the instrument on in set mode. Quality control recommendation Starting the measurement accutrned Close the measurement chamber flap. This starts the actual measurement.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Error messages Error messages In certain circumstances error messages may appear on your display. Quality control recommendation Quality control recommendation To ensure that the instrument is functioning properly, you should regularly use two levels of control solutions for a function check. Also always check the battery condition after the display test.